Brief Introduction to Elitech Data Logger Software:
Our software is mainly used by our users to configure the logging parameters(like temperature unit, logging interval), save or export the data into PDF/Excel format or for other purposes. After our users receive the data logger, they will need to download the software on our site first.
Explanations upon all the four existing softwares:
Elitech Temp Logger Win: For all Windows operation systems, compatible with RC-5, RC-5+, GSP-6, RC-4, RC-4HC, RC-51 & RC-51H data loggers.
Elitech Temp Logger Mac: For Mac operation system, compatible with all temperature data loggers.
Temlog20: For Temlog20 data logger only, this application works on both Windows and Mac system.
Temlog20H: For Temlog20H data logger only, this application works on both Windows and Mac system.
Explanations upon software settings:
After proper downloading and installation, our users will need to configure the parameters in the software before put the logger in use. Please navigate to the Parameter tab, there are a few settings that we need to be aware of:
Log Clock: Not configurable. But the time will sync with that on the computer after our user click the Save Parameter button under the Parameter tab. After that, plug out the logger and then plug in, the time on the software will be updated.
S/N: Not configurable.
Run Time: Not configurable. It depends on the total points of the logger and the log interval, and will change with the interval time.
Address: Configurable. But with no particular meaning, please leave it be.
Trip Description: Configurable. The info entered here will be shown on the report, it can be used to differentiate one logger from another.
Log interval: Configurable. Means how often our user want the logger to perform a reading, it can be 10S, 1minute or 1day.
Start Delay(H): Configurable. Meaning after you press the button on the logger, how long it would wait to officially start logging.
Press button: Configurable. Meaning after we press the triangle button for 4 seconds on the logger to start logging, we can still press this very button to stop the logger.
Temperature Unit: Configurable. Choose the temperature unit we want to see on the logger and on the report, it could be Celsius or Fahrenheit.
High/Low limit: Configurable. The upper and lower alarming point you want to set. After we configure this part and if the logger is with alarming function, each time the temperature is out of the normal range, the logger will give buzzer or LED flashing alarm.
Put the Logger in Use and Export the Data:
After configuring all the parameters, please press the triangle button on the logger and if you see a triangle icon on the screen, it means the logger has been properly started. If you allowed Press Button function in the software, you can stop it by pressing the button again.
After the logger is stopped, plug it in the computer and go to the software. At this moment, the data collected in the logger will be uploaded to the software automatically.
And navigate to the Database tab, double click the data you have uploaded. It will lead you to the Graph tab, there you can export the data the logger has collected to an Excel or PDF format.
After you have saved the data and want to reuse the logger, simply by clicking the Save Parameter button under the Parameter tab, all the data will be cleared. And at this point, you can use the logger again.
Thank you very much for reading this blog, and if you have any more questions and suggestions, please feel free to leave a message below or write to us at firstname.lastname@example.org, we will be very glad to hear from you.